In our newest HeartOS version we have relocated the surface probe calibration settings, which are now found in utilities menu and not within the print calibration settings, as it used to be (as seen in the video https://cellink.sharepoint.com/Delade%20dokument/Forms/AllItems.aspx?id=%2FDelade%20dokument%2FVideos%2FPrinting%20with%20the%20ABL%2Emp4&parent=%2FDelade%20dokument%2FVideos&p=true&slrid=ed4a929e-e0c2-7000-2364-347479dd1dad).
You can follow these steps to calibrate the position of the surface probe.
1. Select Utilities Menu > Machine Calibration
2. Attach the surface probe
3. Jog the surface probe with the XYZ axes until the surface probe is centered in the left well in the printbed (see image below)
4. Select “Calibrate” and you’re all set!